|No. 1,Tech 9th Road,High & New Tech Development Zone,Songshan Lake,DongGuan City,Guangdong Provinceemail@example.com|
Payment & Shipping Terms:
AATCC 201 Drying Rate Tester,
60Hz Drying Rate Tester
HTF-036 Drying Rate Tester-Heated Plate Method AATCC 201
Drying Rate Tester-Heated Plate Method,a fully automated testing machine, is based on the requirement of US AATCC201 testing standard, specially developed an automatic fabric drying rate tester.
Drying Rate Tester, equipped with high precision hot-wire anemometer, IR thermocouple, PT100 temperature sensor, precision water dripping system, precision air flow control system and the air flow funnel designed based on the Fluid Dynamics Theory, together with PC software, automatically finishes a series of jobs, such as the dry time and dry rate tests, and data analysis. It greatly improves the working efficiency and the measurement accuracy, significantly reduces the difference and mistakes which may be introduced by operators.
Data transfer from the main machine to PC through USB cable. Data and curves can be saved and printed by the operators easily.
Wide range of air speed (0.2 – 4.5m/sec) and water dripping amount (0.05 – 4.5ml).
Test Scope:All types of fabrics, including knits, wovens and non-wovens, as well as fabrics taken from end product items.
Fan Box:Generate 1.5m/s air flow
Heated Plate: 305mm × 305mm ± 1mm
Flexible Heater:305mm × 305mm ± 1mm
Insulation Cork Liner:305mm × 305mm ± 1mm
Sample Holder (metal bar): 150mm × 40mm × 2mm ± 1mm
Heated Plate Temperature: 37 ± 0.5 ºC (with over heat protection)
Air Velocity:0.5-4.5m/s ± 0.1m/s
IR thermocouple:15-50ºC ± 0.1 ºC
Water Dripping Range and Accuracy:0.05-4.5ml ± 0.0025ml
Data Transfer:USB cable
Test Interface:Curves and tables
Test Report:Automatic saved and easy print
Contact Person: linmei