|No. 1,Tech 9th Road,High & New Tech Development Zone,Songshan Lake,DongGuan City,Guangdong Provinceemail@example.com|
Payment & Shipping Terms:
|Force::||0-1000N Adjustable||Load Angle Of Armrest::||10 Degrees|
|Seat Height: :||30~65cm||Loading Times: :||0~99999 Times, Adjustable|
|Strength Adjust::||Strength Adjust: Pressure Adjust||Beams Lifting Way::||Electric Lift|
furniture testing equipment,
temperature humidity test chamber,
BIFMA X5.1 Chair Armrest Durability Tester
Furniture Laboratory Equipment For Seating Durability Testing With BIFMA X5.1
TNJ-018 Chair Armrest Durability Testing Machine
Introduction of product:
Chair Armrest Durability Testing Machine use aluminum frame, elegant appearance.
The tester is suitable used for impact testing of all kinds of furniture, such as home furniture, Hotel furniture, restaurant furniture etc. Chair Armrest Durability Testing Machine simulates furniture in process of daily use and Habits misuse, Test the durability of Armrest parts.
|Force: 0-1000N adjustable||0-1000N adjustable|
|Load Angle of armrest:||10 degrees|
|Loading times:||0~99999 times, adjustable|
|Strength adjust:||pressure adjust|
|Control:||PLC, text display|
|Ar supply:||Seat surface height kgf/cm^2 or above|
|Power:||1 AC 220V 50Hz 5A
Standards: BIFMA X5.1-2011-13 Arm Strength Test - Vertical - Static
Reference test method :
13.1 Applicability: This test applies to all chairs with arms.
13.2 Purpose of Test: The purpose of the test is to evaluate the ability of a chair and arm to withstand stresses caused by applying vertical forces on the arm(s).
13.4 Test Procedures
13.4.1 Functional Load
a) A force of 750 N (169 lbf.) shall be applied for one (1) minute.
b) Remove the force.
13.4.2 Proof Load
a) A force of 1125 N (253 lbf.) shall be applied for one (1) minute.
b) Remove the force.
13.5 Acceptance Level
13.5.1 Functional Load
There shall be no loss of serviceability. For a height adjustable arm, failure to hold its height
adjustment position to within 6 mm (0.25 in.) from its original set position as the result of the
loading is considered a loss of serviceability.
Contact Person: linmei