|No. 1,Tech 9th Road,High & New Tech Development Zone,Songshan Lake,DongGuan City,Guangdong Provincefirstname.lastname@example.org|
Payment & Shipping Terms:
|Power:||Electronic||Standard:||EN1888:2012 Section 220.127.116.11.2|
|Product Name:||Laboratory Testing Equipment||Display:||LCD Display, Touch Screen Control|
TC-002B Stroller Durability Test Machine for handle strength(lift up and down)
This machine is suitable for hand-push Baby Carriage beyond the obstacles made by the simulation under
the pressure of raising durability test, and made by CNS test.
Alternately raise and lower the handle(s) by applying a vertical force to the handle so that the rear wheels
and front wheels in turn are raised (120 ± 10) mm, measured at the start of the test from the floor and then
lowered in a controlled manner without pause.
Carry out the test for a total of 10 000 cycles at a frequency of (15 ± 2) cycles/min.
Where the downwards force necessary to lift the front wheels exceeds 450 N, carry out the test by
applying alternately a downwards 450 N force and an upwards force necessary to raise the rear wheels for
3 000 cycles at a frequency of (15 ± 2) cycles/min, then continue the test by only raising the rear wheels
(120 ± 10) mm for additional 7 000 cycles at a frequency of (15 ± 2) cycles/min.
Display: touch-screen, PLC control
Air cylinder stroke: 200~350mm adjustable
Test frequency: 15 ± 2 cycles/min.
Test area: 100x175cm
Beam height adjust: power-driven 35cm
Test station: one position
Grip distance: 11cm to 41 cm
Height of Test wheels from ground: 120±10mm
Test pressure: 450N±5% adjustable
Test cycles: 10000 cycles
Auto counter: 0~99999times, electricity, power cut memory function
Machine dimension: 138x175x200cm
Power: AC ~220V 50Hz
Weight: About 450KG
Standard: EN1888:2012 section 18.104.22.168.2