|No. 1,Tech 9th Road,High & New Tech Development Zone,Songshan Lake,DongGuan City,Guangdong Provinceemail@example.com|
1.Purpose for toys impact tester：Use impact strength test methods to simulate a child playing or abusing toys or other items to determine the possibility of injury.
2.Toys impact tester apparatus including :
2.1 level steel 1 pcs
2.2 Metal weights - 1 ± 0.02kg weight, diameter:80 ± 2mm
2.3 Scale – measured range 200mm.
3.Test Method for toys impact tester:
3.1.Specimen prepare: Before the test, each sample run should be placed at such environment condition of a temperature 20-25 ℃, relative humidity 20 ~ 70% for at least 4 hours.
3.2.Procedure for toys impact tester:
3.2.1. Put the weakest point of the sample on the steel surface,doping the metal weights on the test sample from 100 ± 2mm height .
3.2.2. Repeat the test as 3.2.1 one time more.
3.2.3.Check test sample whether it has the small objects, sharp corners, sharp edges or any dangerous parts.
4. Standard Reference for toys impact tester: EN71-1 8.7 section
Dongguan Hust Tony Instruments Co.,Ltd. focus on design and manufacturing the lab testing machines including furniture testing equipment,toys testing machines, all the testing machines are designed and produced base on the standards of BIFMA、ASTM、DIN、EN71、GB、ISO、AATCC etc. We supply the complete solution of laboratory testing.
Contact Person: Mr. Lewis Huang